AEL single column vertical machine instruction manual
The AEL single column vertical machine is a push-pull load test bed for the HF series and NK series push-pull gauges.This product adopts double-pole structure, which has the advantages of good stability, wide application range, convenient use, etc. It also has the function of stepless speed regulation, manual (inching) and automatic control switching.Applicable to rubber and plastic, light industrial textile, building doors and windows, composite materials, wire and cable, auto parts, power machinery, scientific research institutions and other industries push-pull load, insertion force, destruction test and other tests.
1. Electric loading, stepless speed regulation, gear transmission
2, compact structure, smooth transmission
3, simple operation, and can use a variety of fixtures
The specification parameters:
Dynamometer installation method:
Built-in sensor: Insert the 4-M6 screw into the 4-φ6.2 hole of the fixing block, insert a washer in the thread direction, and tighten the 4-M6 thread on the dynamometer mounting plate. The dynamometer is installed.On the board, the dynamometer mounting holes are divided into two groups of 30x145 and 40x90, respectively.Users can determine according to their own needs.
The use of control panel:
1. Power switch: When the button is pressed, turn on the power and press it again to turn off the power.
2. Speed control: Due to the difference between the measured object and the user's requirements, the speed of the carrier is 20 to 200 mm/min when adjusting the governor knob.
Manual/automatic, when turning the knob to manual, if you want to move the carrier, you need to press and hold the up or down key. When you turn off, the movement stops.When the knob is turned to automatic, if the carrier is moved, press the up or down key directly.
There are two screws on the rear side of the column, which is loosened with the standard hexagon wrench and adjusted to the proper position to control the sliding block.
Counter use method:
The top of the counter is a digital display, recording the number of trials, the bottom is a preset button, a total of four, from left to right one, ten, hundred, thousand, indicating that the preset can be set in the range of 0 to 9999.For example, if the user conducts a continuous test on a certain product, the number of tests is 9999, which needs to be adjusted on four buttons respectively, and the display is changed to 9999. Press the manual/automatic button again to test 9999 times and run 9999 times automatically.Stop. If you need to test again, press the stop button.
1. How can I choose the suitable one?
Dear customer, please tell us your detailed requirements by mail or online, we will recommend the suitable one as your request.
2. Does your price is competitive?
Dear customer, we make sure to offer you the best quality with competitive price.
3. How can I pay?
Dear customer, we accept many payment term, such as T/T, Western Union…
4. When I receive it after pay?
Dear customer, normal models can be delivered with 5-7days, please contact us to check transport time to your address.
5. How to deliver?
Dear customer, we can send by Express, by Sea and by Air.
6. Is will be broken during transport?
Dear customer, please do not worry, we do standard export package.
7. What should I do if I do not know how to use?
Dear customer, please do not worry, manual user will be sent together, you can also contact us with more technological support.
8. What should I do if some parts broken?
Dear customer, please do not worry, we have 12months warranty except wear parts. You can also buy p